The AirCurve™ 10 S BiLevel Machine with HumidAir™ Heated Humidifier by ResMed is a cutting-edge unit designed to offer comfort through intuitive technology. Building on earlier versions, this bilevel machine delivers one pressure for inhalation and one pressure for exhalation. The heated humidifier is built-in to the machine to provide moisture and warm air through the hose into the mask.
Res Med AirCurve S10 Bi-Level Machine with HumidAir Humidifier
The AirCurve™ 10 S BiLevel Machine with HumidAir™ Heated Humidifier operates on a set, higher pressure for inspiration and a set, lower pressure for expiration. The machine’s comfort features are designed to help the user adjust to the two delivered pressures. This is a bilevel machine delivering one pressure for inhalation and one for exhalation while monitoring the length of the breaths and responding to alter the timing as needed to maintain the respiratory cycle.
- Easy-Breathe Comfort Feature
- Built-In HumidAir™ Heated Humidifier
- TiControl and Adjustable Rise Time
- LCD Displays with My Options
- Enhanced Vsync Technology
- SmartStart / SmartStop
- SD Card and Data Storage
Easy-Breathe Comfort Feature
The Easy-Breathe feature on the AirCurve™ 10 S is designed to imitate a more natural breathing pattern. At the point of transition between the two pressures, a waveform of increasing and decreasing pressure is released to encourage a natural resting breath.
Built-In HumidAir™ Heated Humidifier
The AirCurve™ 10 S has a fully integrated humidifier built-in to the machine to supply moisture and a warmer temperature throughout the hose to increase therapy comfort. Using the My Options screen on the machine, the humidification level is easy to set. Numbers 1 through 8 are listed on the screen, 1 represents the lowest level of humidification, and 8 represents the highest. The humidifier is not required for therapy, so it may be turned off if it is not desired.
The addition of the OPTIONAL ClimateLineAir™ heated tube assists in maintaining the warm temperature throughout the hose to prevent Rainout as the environmental temperature changes.
TiControl and Adjustable Rise Time
The TiControl feature on the AirCurve S machine monitors the pattern of inhalation and exhalation, comparing it against the TiMax and TiMin settings on the machine. Those with respiratory disorders such as COPD or obstructive lung disease may experience early or delayed changes from inspiration to expiration, referred to as cycling. The 10 S BiLevel TiControl feature is designed to prevent cycling by setting a minimum and maximum time limit on inspiration. When the machine detects the user breathing which falls outside the machine settings, it will intervene to alter the timing of the delivered IPAP (inhalation) or EPAP (exhalation) pressure.
- Ti Max: Maximum length of Inhalation
- Ti Min: Minimum length of Inhalation
- Trigger: Sensitivity level for inhalation trigger
- Cycle: Sensitivity level for exhalation
Adjustable rise time is similar to TiControl. Both features support the user’s ability to breathe at the appropriate time. Rise time is designed for this bilevel machine to increase the range of time from expiration to inspiration based on the set time requirement. Please Note: The rise time should not exceed the time set for inspiration due to a required higher pressure at inhalation.
IMPORTANT FUNCTION NOTE: The AirCurve 10 S machine does spontaneously regulate the breathing pattern to avoid cycling which results in inefficient breathing. This machine DOES NOT deliver a breath on a timed basis. This machine is NOT a BiLevel ST machine and DOES NOT have a back up rate setting.
LCD Display with My Options
The AirCurve™ 10 S BiLevel has a color LCD display. The menus are designed to be easy to read and use. Select the menu item on the screen by turning the easy grip dial. Select the My Options menu to view and modify the current settings:
- Ramp time – Ease into therapy with a period of lower pressure, or select AutoRamp™ allowing the machine to sense and adjust when sleep begins.
- Humidity Level – Set humidity between 1 (lowest) & 8 (highest), or turn it off.
- Mask Fit – Runs a seal integrity check showing a green smiling face for a good fit or red frowning face for a poor fit.
- Leak Alert- Set the device to begin beeping when a large leak is detected or the mask is removed.
- SmartStart™ – Therapy automatically starts when it senses breathing.
- SmartStop™ – Therapy automatically stops when the mask is removed.
- Mask Type – Set the mask type being used: Nasal; Nasal Pillow; Full Face Mask.
- Run Warmup – Preheat the humidifier water before starting therapy.
Enhanced Vsync Technology
If changes occur in mask pressure or the seal of the mask, Vsync technology is established to quickly detect the leak. Vsync leak management is developed by ResMed to track and control leakage and to relay accurate data in the instance that a leak arises. The goal is to provide the user with a factual estimation of airflow. Likewise, the differences in lung capacity can be determined by the Trigger and Cycle Sensitivity feature. The level selected reflects the respiratory effort of the user.
SmartStart™ / SmartStop™
The intuitive AirCurve™ 10 S SmartStart™ component automatically starts the device when the user breathes into the mask. The SmartStop™ component stops the device when the user removes the mask.
SD Card and Data Storage
The AirCurve™ 10 S BiLevel records the information about the sleep therapy, which can be downloaded onto an SD Card. The SD Card is installed in the machine at the time of shipment and must be in the machine before it is used. The machine’s on board memory holds compliance information for 365 sessions of use as well as detailed data for 30 sessions.
CPAP Video & FAQ
This Product Includes…
- BiLevel Machine
- Built-In HumidAire Heated Humidifier
- Standard Water Chamber
- 6 Ft SlimLine Tubing – 15mm Diameter
- Power Supply and Cord
- Travel Bag
- 1- Standard Filter (Installed)
- Filter Cover
- SD Card – Inserted in Machine
- Patient Manual
Weight – Machine with Water Chamber: 2.75 lbs (44 oz.)
Dimensions – Machine with Water Chamber: 10.04 in L x 4.57 in W x 5.91 in H
Weight – Standard Chamber: .54 lb
Weight – Machine with Side Cover (Without Water Chamber): Approx. 2.452 lbs
Dimensions – Machine with Side Cover (Without Water Chamber): Approx. 8 in L x 4.57 in W x 5.91 in H
Dimensions – Travel Bag: 13.24 in L x 10 in W x 5 in H
Weight – Travel Bag Alone: 1.18 lb
Weight – Machine with Humidifier; Power Supply and Cord; SlimLine Tube; Travel Bag: Approx. 5.1 lbs
Weight – Machine with Humidifier; Power Supply and Cord; OPTIONAL ClimateLineAir Tube; Travel Bag: Approx. 5.25 lbs
Sound Level (CPAP Mode): 26.6 dBA
Water Chamber Capacity: 380 mL
Weight – 90W Power Supply, Cord and Plug: 1.29 lb
Dimensions – 90W Power Supply Only (Not including Cord or Plug): 5 1/2 in L x 1 7/8 in W x 1 1/4 in H
AC Input Range – 90W Power Supply: 100 – 240V, 50 – 60Hz, 1.0 – 1.5A, Class II
DC Output – 90 W Power Supply: 24V — 3.75A
Typical Power Consumption: 53W (57VA)
Peak Power Consumption: 104W (108VA)
Operating Pressure VPAP Mode: 4-25 cm H2O
Operating Pressure CPAP Mode: 4 – 20 cm H2O
Materials – Water Chamber: Injection Molded Plastic, Stainless Steel, Silicone Seal
Materials – Standard Filter: Polyester Non-woven Fiber
Included Hose Dimensions: 6 Ft Long, 15mm Interior Diameter, 22mm Cuff attachment
Hose Materials: Flexible Plastic
Temperature Range – Optional ClimateLineAir™ Heated Tube: 60 – 80° F (16 – 30° C)
Operating Temperature: 41° F – 95° F (5° C – 35° C)
Storage and Transport Temperature: -4° F – 140° F (-20° C – 60° C)
Operating Humidity: 10% – 95% Relative Humidity, non-condensing
Storage and Transport Humidity: 5% – 95% relative Humidity, non-condensing
Operating Altitude: Sea Level – 8500 Feet (2,591 M); Air pressure range 1013 hPa – 738 hPa
Maximum Temperature – Heating Plate: 154° F (68° C)
Maximum Temperature – Cut Out: 165° F (74° C)
Maximum Gas Temperature: Less than or Equal to 106° F (41° C)
IEC 60601-1:2005 Classification: Class II (Double Insulation), Type BF, Ingress protection IP22
Machine and Humidifier Warranty: 2 Years
Patient Weight: Greater than 66 lbs
Optional Supplemental Oxygen: Operates with up to 15L/min of supplemental oxygen
The chamber should be taken apart and hand washed weekly using warm water and a mild detergent. It should be rinsed well and allowed to air dry out of direct sunlight before reassembling. The chamber SHOULD NOT be washed in a dishwasher or washing machine. The long hose should be washed in the same manner and allowed to completely dry out of direct sunlight before reattaching to the machine and mask.
Required Settings: This machine requires two pressure settings and a Spontaneous rate.
Unless otherwise noted, CPAP.com will leave the AirCurve™ 10 S BiLevel Machine at the following default settings:
- Ramp Time – 20 min
- Start EPAP – 3.0
- Humidity Level – 4
- Mask – Full Face
- Airplane Mode – OFF
- Mode – S
- Easy-Breathe – ON
- Ti Max – 2.0s
- Ti Min – 0.3s
- Rise Time – Min
- Trigger – Med
- Tube – Slimline
- AB Filter – No
- Leak Alert – OFF
- SmartStart – OFF
- Language – English
- Temperature – Degree Fahrenheit
- Some product images supplied by ResMed © ResMed 2015 Used with Permission.
- ResMed products are for sale only to patients within Canada and not to other third parties or businesses.