TheThe XT-III is a fixed pressure CPAP device used for the treatment of OSA. This product includes an easy-to-use, menu-driven LCD user interface. Data is accumulated on the XT-III can be transferred onto a PC, where the caregiver can review and evaluate patient usage data, including compliance information, daily pressure charts and statistical reports.
This product includes an easy-to-use, menu-driven LCD user interface. Data is accumulated on the XT-III can be transferred onto a PC, where the caregiver can review and evaluate patient usage data, including compliance information, daily pressure charts and statistical reports.
Enhanced pressure sensor for even better pressure accuracy
Automatic leak adjustment and altitude compensation
Data Management Software can be used to downlod data to a PC
Built-in alarm clock
Alert notification for tube change
This Product Includes…
XT CPAP Machine
6 Foot Hose
Carrying Case with Shoulder Strap
1 – Reusable Foam Filter (Installed)
1 Year Warranty
Sound Level: 28 cBA at 10 cmH2O, 1 Meter Distance
Machine Alone Dimensions: 5.7 in x 5.1 in x 3.9 in (14.5cm x 13cm x 10cm)
Weight – Machine Alone: Approximately 1.87 lbs
Weight – Machine with Power Cord: Approximately 2.2 lb
Dimensions – Smaller Carry Bag (If Purchasing Machine Alone): 11″ L x 7″ W x 7″ H
Dimensions – Larger Carry bag (If Purchasing Machine with Humidifier): 13″ L x 7.5 ” W x 8 ” H
Weight – Machine with Smaller Carry Bag & Power Cord: Approximately 3.3 lbs
Weight – Machine with Larger Carry Bag, Power Cord, and OPTIONAL Heated Humidifier: 5.8 lbs
Weight – Smaller Carry Bag Alone: Approximately 1.1 lbs
Weight – Larger Carry Bag Alone: Approximately 1.3 lbs
Power Supply: AC100-240V, 50/60 Hz, 0.5-03A
Pressure Range: 4-20 cmH2O (Adjustable in 0.5 cmH2O Increments)
Initial Pressure: 3-19 cmH2O (Adjustable in 0.5 cmH2O Increments)
Maximum Pressure: 5-20 cmH2O (Adjustable in 0.5 cmH2O Increments)
Minimum Pressure: 4-19 cmH2O (Adjustable in 0.5 cmH2O Increments)
Ramp Time – CPAP Mode Only: 0-45 Minutes (Adjustable in 5 Minute Increments – CPAP Mode Only)
Ramp Starting Pressure – CPAP Mode Only: 3-19 cmH2O (Adjustable in 0.5 cmH2O Increments)
Operating Altitude (Auto Altitude Adjust):
Pressure of 4 -18 cmH2o = Up To 8,000 Feet
Pressure of 18.5 – 20 cmH2o = Up To 5,000 Feet
Summary Data – 730 Sessions
Efficacy Data (AHI, Leak) – 18 Sessions or Cumulative 72 Hours
Operating Conditions: Temperature 41°F to 95°F (5°C to 35°C); Non-Condensing Humidity 15% to 95%
Storage Conditions: Temperature 5°F to 122°F (-15°C to 50°C); Non-Condensing Humidity 10% to 90%
Shipping Conditions: Temperature 5°F to 122°F (-15°C to 70°C); Non-Condensing Humidity 10% to 90%
Required Settings: This machine does not have required settings.
Unless directed otherwise, CPAP.com will leave this machine at its default settings of:
PVA Pressure Relief: Off
Clock Alarm: Off
When Auto Mode is Selected: 4 – 20 cm H2O is Default Pressure Range
Auto Altitude Adjustment: On (Cannot be Disabled)
The device should not be operated near flammable anesthetic mixtures in combination with nitrous oxide and oxygen.
Airflow from the device may be as much as 12.6° F (7° C) higher than ambient room temperature.
Do not use in ambient room temperatures higher than 95° F (35° C).
The device will stop providing airflow and provide an error message if it overheats. The machine can be restarted once it has cooled.
The device should not be used if the power cord or plug is damaged.
The power cord should be kept away from hot objects and surfaces.
The air outlet of the device should never be blocked or placed in an area where they can become blocked.
The air outlet of the device should be kept free of debris such as lint, hair, and other particles that could obstruct the air outlet.
If disposing, region or country regulations must be followed.
DO NOT allow the machine to be submerged in water or to be stored or operated in a place where it could be pulled or fall into water.
DO NOT retrieve the device if it has come into contact with water and is still plugged in. The device should be unplugged immediately.